a Department of Cardiac Surgery, University Heart Centre Zurich, University Hospital Zurich, Switzerland
b Department of Cardiology, Electrophysiology and Cardiac Devices, University Heart Centre Zurich, University Hospital Zurich, Switzerland
c HerzKlinik, Klinik Hirslanden, Zurich, Switzerland
We report a modified surgical access for subcutaneous implantable cardioverter defibrillator implantation to preserve an artistic tattoo providing uncompromised device positioning and function in a 35-year-old patient with Brugada syndrome.
A 35-year-old male patient was referred from an outside hospital after surviving a sudden cardiac arrest. Subsequent cardiac assessment revealed Brugada syndrome and implantation of an implantable cardioverter defibrillator (ICD) for secondary prevention was planned. As there was no need for cardiac pacing in the near future, a subcutaneous ICD (S-ICD) was chosen. Pre-interventional diagnostics demonstrated that all three vectors for the S-ICD fulfilled the screening criteria.
However, at the site of the device pocket the patient had an artistic cutaneous tattoo showing a clock with Roman numbers. Surgery was performed with an interdisciplinary approach by a cardiologist and cardiac surgeon with the particular intention to preserve the tattoo. In order to minimise the risk for optical distortion of the clock, the device pocket access was anteriorised and the cutaneous incision was made at the anterior margin of the tattooed clock by a semi-circular incision on the level of the anterior axillary line. Despite this unusual anterior access, subcutaneous blunt preparation of the inter-muscular device pocket between the latissimus dorsi and anterior serratus muscle was uneventful (fig. 1A). Hence, the device could be placed in the correct position providing an optimal vector for defibrillation with the parasternal subcutaneous lead. After fixation of the device to the serratus anterior fascia, the pocket was closed layer by layer and an intra-cutaneous suture using Maxon 4/0 along the anterior margin of the clock was performed to close the wound (fig. 1B). Post-implant device testing revealed an uncompromised device function and radiographic imaging confirmed a correct position of the ICD (fig 1D, E).
During post-interventional follow-up 4 weeks after surgery the patient had no arrhythmic events or ICD shocks and an uncomplicated wound healing with an aesthetic cicatrisation without optical distortion of the filigree tattoo (fig. 1C).
This case demonstrates successful optical preservation of an artistic tattoo by a modified surgical access for S-ICD implantation without restriction of neither an optimal device position nor its function.
Dr. Steffel has received consultant and / or speaker fees from Abbott, Alexion, Amgen, Astra-Zeneca, Bayer, Berlin-Chemie, Biosense Webster, Biotronik, Boehringer-Ingelheim, Boston Scientific, BMS, Daiichi Sankyo, Medscape, Medtronic, Merck/MSD, Organon, Pfizer, Saja, Servier, and WebMD. He reports ownership of CorXL. Dr. Steffel has received grant support through his former institution (University Hospital Zurich) from Abbott, Bayer Healthcare, Biosense Webster, Biotronik, Boston Scientific, Daiichi Sankyo, and Medtronic.
Dr. Breitenstein has received consultant and / or speaker fees from Abbott, Bayer Health Care, Biotronik, BMS/Pfizer, Boston Scientific, Daiichi Sankyo, and Medtronic. Dr. Breitenstein has received educational grant from Biosense Webster, Biotronik and Actelion. Dr. Breitenstein has received speaker honoraria from Abbott, Biotronik, BMS/Pfizer, Bayer Health Care, Abbott, Spectranetics/Philipps, and Medtronic
Dr. Saguner has received a speaker honorarium from Boston Scientific and educational grants from Abbott, Biotronik, BMS/Pfizer and Boston Scientific.
Devdas Thomas Inderbitzin, MD
Department of Cardiac Surgery
University Hospital Zurich
Published under the copyright license
“Attribution – Non-Commercial – NoDerivatives 4.0”.
No commercial reuse without permission.